All information products included in http://ics-cert.us-cert.gov are provided "as is" for informational purposes only. The Department of Homeland Security (DHS) does not provide any warranties of any kind regarding any information contained within. DHS does not endorse any commercial product or service, referenced in this product or otherwise. Further dissemination of this product is governed by the Traffic Light Protocol (TLP) marking in the header. For more information about TLP, see http://www.us-cert.gov/tlp/.
1. EXECUTIVE SUMMARY
- CVSS v3 5.3
- ATTENTION: Exploitable remotely/low skill level to exploit
- Vendor: GE
- Equipment: Aestiva and Aespire Anesthesia
- Vulnerability: Improper Authentication
2. RISK EVALUATION
Successful exploitation of this vulnerability could allow an attacker the ability to remotely modify GE Healthcare anesthesia device parameters. This results from the configuration exposure of certain terminal server implementations that extend GE Healthcare anesthesia device serial ports to TCP/IP networks.
3. TECHNICAL DETAILS
3.1 AFFECTED PRODUCTS
The following versions of GE Aestiva and Aespire Anesthesia Machines, are affected:
- GE Aestiva and Aespire Versions 7100
- GE Aestiva and Aespire Versions 7900
3.2 VULNERABILITY OVERVIEW
A vulnerability exists where serial devices are connected via an added unsecured terminal server to a TCP/IP network configuration, which could allow an attacker to remotely modify device configuration and silence alarms.
- CRITICAL INFRASTRUCTURE SECTORS: Healthcare and Public Health
- COUNTRIES/AREAS DEPLOYED: United States
- COMPANY HEADQUARTERS LOCATION: United States
Elad Luz of CyberMDX reported this vulnerability to NCCIC.
GE Healthcare recommends organizations use secure terminal servers when connecting GE Healthcare anesthesia device serial ports to TCP/IP networks. Secure terminal servers provide robust security features, including strong encryption, VPN, authentication of users, network controls, logging, audit capability, and secure device configuration and management options.
GE Healthcare recommends that organizations utilize best practices for terminal servers that include governance, management, and secure deployment measures such as network segmentation, VLANs, and device isolation to enhance existing security measures.
GE Healthcare plans to provide updates and additional security information about this vulnerability for affected users at the following location:
NCCIC recommends users take defensive measures to minimize the risk of exploitation of this vulnerability. Specifically, users should:
- Minimize network exposure for all medical devices and/or systems.
- Locate medical devices behind firewalls and isolate them where possible.
- Restrict system access to authorized personnel only and follow a least privilege approach.
- Apply defense-in-depth strategies.
- Disable any unnecessary accounts, protocols and services.
- Where additional information is needed, refer to existing cybersecurity in medical device guidance issued by the FDA at the following location:
NCCIC reminds organizations to perform proper impact analysis and risk assessment prior to deploying defensive measures.
NCCIC also provides a section for control systems security recommended practices on the ICS-CERT web page. Several recommended practices are available for reading and download, including Improving Industrial Control Systems Cybersecurity with Defense-in-Depth Strategies.
Additional mitigation guidance and recommended practices are publicly available on the ICS-CERT website in the Technical Information Paper, ICS-TIP-12-146-01B--Targeted Cyber Intrusion Detection and Mitigation Strategies.
Organizations observing any suspected malicious activity should follow their established internal procedures and report their findings to NCCIC for tracking and correlation against other incidents.
No known public exploits specifically target this vulnerability.
For any questions related to this report, please contact the NCCIC at:
Toll Free: 1-888-282-0870
The NCCIC continuously strives to improve its products and services. You can help by choosing one of the links below to provide feedback about this product.